|SKU||Description||Volume Range||Volume Increment||Accuracy||Precision||Quantity||Price||Add to Cart|
|712111019999||MicroPette Single Channel Variable Volume Pipettor||0.1 to 2.5 μL||0.05 μL||±3.0% (1.25 μL)||±3.0% (1.25 μL)||1 Each||
|712111049999||MicroPette Single Channel Variable Pipettor||0.5 to 10 μL||0.1 μL||±1.5% (5 μL)||±1.5% (5 μL)||1 Each||
|712111089999||MicroPette Single Channel Variable Pipettor||10 to 100 μL||1.0 μL||±1.0% (50 μL)||±0.4% (50 μL)||1 Each||
|712111059999||MicroPette Single Channel Variable Volume Pipettor||2 to 20 μL||0.5 μL||±1.2% (10 μL)||±1.0% (10 μL)||1 Each||
|712111069999||MicroPette Single Channel Variable Volume Pipettor||5 to 50 μL||0.5 μL||±0.9% (25 μL)||±0.6% (25 μL)||1 Each||
|712111099999||MicroPette Single Channel Variable Volume Pipettor||20 to 200 μL||1.0 μL||±0.8% (100 μL)||±0.3% (100 μL)||1 Each||
|712111149999||MicroPette Single Channel Variable Pipettor||100 to 1000 μL||5.0 μL||±0.7% (500 μL)||±0.25% (500 μL)||1 Each||
|712111179999||MicroPette Single Channel Variable Pipettor||1 to 5 mL||50.0 μL||±0.6% (2500 μL)||±0.3% (2500 μL)||1 Each||
|712111339999||MicroPette Single Channel Variable Pipettor||2 to 10 mL||100.0 μL||±1.2% (5000 μL)||±0.3% (5000 μL)||1 Each||
Eight variable volume MicroPette's in the range. Volume adjustment is selected by turning the thumb plunger mechanism. MicroPette's are calibrated in accordance ISO/DIS 8655. Each pipettor passes a quality control/calibration test and is supplied with a certificate.
- Pipettes cover a volume range from 0.1μL to 10mL
- Ergonomic design provides excellent operating experience
- Large display window allows for easy volume identification
- Integral tip ejector for access into narrow tubes
- Easy to calibrate and maintain with tool supplied
- Autoclavable tip-cone
- Used with 'universal' pipette tips
- Each MicroPette supplied with an individual calibration certificate according to ISO8655
- 1 YEAR WARRANTY
The MicroPette Plus pipettes are fully autoclavable, enable easy cleaning and reduce the risk of contamination. Steam autoclaving can be performed at 121°C, 1 bar for 20 minutes. After autoclaving, the pipette must be cooled down and left to dry for 12 hours before use.
It is recommended to check the performance of the pipette after each autoclaving. Greasing and sealing of the pipette piston after every 10th autoclaving will secure an enhanced function.
All SCILOGEX pipettes have been quality tested according to ISO8655-2:2002 and are supplied with individual calibration certificates. The quality control includes gravimetric testing of each pipette with distilled water at 22°C.